There are so many motors on the market today, from heavy-duty AC motors to tiny DC brushless and stepper motors. To choose a motor, users must have a full understanding of the application parameters, including power, speed, torque, physical size, efficiency, lifetime expectations, and other requirements. There have literally been entire books published about each of these aspects, and to define them all accurately in a short article would be difficult, to say the least.
What this article will do is spell out the primary differences between DC brushed and brushless motors, what they can do, where they fit best, and how to decide which to use in your application. DC motors are being used in more and more industrial applications because of their flexibility and long life. Therefore, DC motor selection is often one of the most important steps in providing motion control of a medical machine, whether for prosthetics, medical tools, robotics, or medical training aids.
Defining the Application
The medical market is unique in that most devices and machines are operated in a clean environment and around people as opposed to an industrial application. This means that long life and low noise are key characteristics the drives need to have in order to fit the requirements of many medical machines. DC motors are noted for their life-spans, particularly DC brushless motors, which can last tens of thousands of hours in continuous operation, and much longer when operated intermittently.
Key criteria for selecting a DC motor for a medical machine application includes finding out what voltage is readily available for the application and what physical size the motor needs to be. Speed and torque can be determined once these two parameters are determined.
Voltage availability is a critical element in motor selection. Prosthetics, for example, are battery operated, while many rack-mounted devices and surgical tools operate from a 24V power supply. DC motors are available for use at voltages as low as 1.5V and as high as 48V dependent on required power.
Physical size is often one of the limiting factors in motor selection for medical machines. Often a compromise needs to be made between which motor to use and the available space it needs to fit into. The prosthetics talked about earlier would need a small frame motor, while rack-mounted devices can be designed to accommodate larger devices.
Efficiency becomes a primary concern when you need to worry about power consumption to maximize battery life in a prosthetic hand or in a portable surgical tool like a drill or saw. Such concerns are not so evident in robotic machines used to perform many surgeries today.
As mentioned before, torque and speed also have an effect on motor frame size. High torque motors are often larger in size than their low-torque counterparts, which means that larger mounting hardware and larger housings may be a requirement of the machine. For example, it takes a larger motor to rotate the magnets in an MRI than it does to run an infusion pump for drug delivery.
Motor duty cycle could be one of the most telling aspects of a medical machine. Intermittent operation not only reduces the wear and tear on the motor, and increases the life of the motor, it also means that a smaller motor size can be used without depleting the positive characteristics of the machine itself.