Fine thin metal laser cutting is an ideal technology for the specialized cutting requirements found in the manufacturing of medical tube tools and components, which require an array of cut features with sharp edges, contours, and patterns within edges. From surgical instruments used in cutting and biopsy, to needles containing unusual tips and side wall openings, to puzzle chain linkages for flexible endoscopes, laser cutting provides higher precision, quality, and speed than traditionally used cutting techniques.
Laser Cutting Benefits
The laser is ideal for medical cutting, because the laser can be focused down to a 0.001 inch diameter spot size that offers a fine non-contact “tool-less” cutting process at high speed and high resolution. As the laser cutting tool does not rely on touching the part, it can be oriented to make any shape or form, and used to make unique shapes.
For example, laser cutting is an excellent tool for small tubes, like those used for cannula and hypo tube applications that require an array of features, such as windows, slots, holes, and spirals. With a focused spot size of 0.001 inches (25 microns), the laser offers high-resolution cuts that remove the minimal amount of material to enable highspeed cutting according to the dimensional accuracy required.
Also, since laser processing is noncontact, no mechanical force is imparted onto the tubes—there is no push, drag, or other force that might bend a part or cause flex that would have a negative impact on process control.
The laser can also be precisely set during the cutting process to control how hot the work area gets. This is significant, because the size of medical components and the cut features is shrinking, and small parts can heat up quickly and might otherwise overheat.
Laser Cutting Mechanism and Equipment
Figure 1 shows the key elements of the fiber laser with gas assist typically used for fine cutting for medical components. A highly focused laser is used to melt a thin sliver of material. While the material is still molten, a 0.02 inch diameter gas jet nozzle that is coaxial with the laser blows away the molten material. The desired features are produced using this continual cycle of melt, then melt ejection. Distance between the laser and the material needs to be maintained precisely. Figure 2 shows a closer view of the laser cutter.
Material Selection and Thickness
Laser cutting is very well suited for the vast majority of materials relevant to potential medical cutting applications, including stainless steels (300 and 400 series, 17-4, 17-7), MP35N (cobaltchrome steel alloy), and Nitinol. Other materials, such as titanium for implants and platinum iridium (Pt-Ir) commonly used for electrodes, can also be laser-cut, though these materials require a little more care in the setup. With titanium, thermal build up and runaway are concerns that can be mitigated by using pulsed laser output that finely controls part heat input and using a semi or fully inert assist gas. For Pt-Ir, excessive dross on the underside of the cut is the main issue. This can also be improved with optimized pulsed laser cut parameters.
The laser can cut materials with a wide range of thicknesses and can be operated in a variety of cutting conditions to achieve cuts a different material thickness. In virtually all medical cutting applications, the laser is operated in pulse mode due to the requirement for fine feature sizes and high tolerance parts. Considering a typically sized laser of 100W, the duration and frequency of the pulses determine the speed of the cut and also the maximum thickness of the cut. This is also related to the feature size—the finer the feature size, the slower and less thick the cut can be, though this is application specific. A 100W laser can comfortably cut a range of stainless steels up to 0.03 inch thick simply by altering the pulse width and frequency. The thinner the material, the shorter the pulse duration and the higher the pulse frequency. For instance, for 0.008 inch thick material, a pulse width of 50 microseconds and a frequency around 3,000 Hz is a good starting point.
The laser excels in tube cutting singlesided features such as holes and slots, offaxes geometries, and spirals that are typical features for cannula and hypo tubes. When coupled to an appropriate motion system and integrated laser control, the laser cutting provides a high throughput system for many standard cutting profiles as well as unique profiles. With an effective cut width of around 20 to 30 microns, it is possible to achieve very high dimensional control, routinely to ±0.001 inch. When cutting accuracy is less fine, cutting speeds can be rapid.
Cutting an edge that is a 90 degree angle to the part may be sufficient for many medical tube applications, but some applications require cutting parts at an angle other than 90 degrees. Laser cutting offers such off-axes cutting, where cut edges are angled to the surface, and can also accommodate compound angle cutting.
Laser Cutting Limitations
As with any process, laser cutting is subject to certain limitations, primarily cutting certain angles in certain orientations. This relates to part thickness, cut quality and the mechanical interference of the equipment. For example, certain designs of arthroscopic surgery shaver blades require a cut angle of up to 70 degrees to the distal part of the tube. The wall thickness can be up to 0.015 inch, which requires a through cut penetration of 0.044 inch. Perhaps more importantly, increasing the angle makes the focus spot more of an ellipse shape, reducing the concentration of laser power. This power density reduction, coupled with the requirement to cut a thicker section, leads to a breakdown of process due to insufficient power density required for the increased thickness.
The laser head can become impeded due to mechanical interference. In some instances, the cutting nozzle may interfere with the tube for high angle cuts. Cut angles of up to 30 degrees to the material’s surface are possible; with certain material thickness conditions and required cut quality, it may be possible to attain angles up to 45 degrees, but anything above that is not attainable at this time.